» Residential Electrical Wiring: Guide to Home Wiring
» Need Electrical Help? Get a Quick Reply! Ask an Electrician
Electrical Testing for Stray Voltage
By Dave Rongey
Summary: This article was originally intended for industrial applications but can be a helpful resource when using digital electrical testers for home applications as well as it applies to testing for stray electrical voltage.
How Stray Voltage Affects Multimeter Measurements
How Stray Voltage Affects Multimeter Measurements
...and What You Can Do About It.
This article was originally intended for industrial applications but can be a helpful resource when using digital electrical testers for home applications as well.
For most electrical measurements in the industrial environment, a high impedance digital multimeter or electrical tester is the appropriate tool. These test instruments have a relatively high input impedance (>1 megohm), which means that when connected, they don’t load the circuit under test. These test tools generally will not affect circuit operation or circuit measurements when testing for stray electrical voltage.
On the other hand, low impedance test instruments can seriously load down a circuit under test and, in some cases, can adversely affect circuit operation and circuit measurements. This can happen with sensitive control circuits or in many electronic circuits found within industrial machinery.
However, even with a high impedance multimeter, there’s still one confusing measurement situation that can occur in facilities or manufacturing plants. It’s called stray or ghost voltage, and it happens specifically when using a high impedance test instrument for day to day measurements.
Stray or ghost voltages occur from capacitive coupling between energized circuits and non energized, non connected adjacent wiring. Because of this coupling effect and the multimeter’s high impedance, it’s not always possible to determine if the circuit under test is energized or de-energized, and this creates confusion for the person performing the test.
Where are stray voltages encountered?
The most common place to encounter stray voltage is in unused cable runs or electrical wiring in existing conduit. When facilities or buildings are built and wired, it's very common for electricians to pull extra wire through the conduit for future use.
These wires are typically left unconnected until needed, but are subject to capacitive coupling from the powered wires.
Another example is an open ground or neutral on a 120 V branch circuit or in card cages where 120 volt control circuits are used to control assembly line or conveyor functions.
What does stray voltage look like?
Is this voltage real? Yes, it is, but it’s a static voltage, containing no real energy or current flow. When it comes to determining whether a circuit or connection is energized, this stray voltage reading presents a real source of confusion. Is the connection really hot or not?
The Fluke TL225 Stray Voltage Adapter Test Lead Set
The Fluke Stray Voltage Adapter Test Lead Set is an accessory that allows a high impedance multimeter to measure circuits, connections, cables or connectors subject to stray voltages. The adapter provides a low impedance load to the measured circuit, desensitizing the meter to low energy, spurious sources of interference. If the measurement points are energized with a hard voltage, the meter will simply display the voltage reading. If the measurement points contain a stray or ghost voltage, the meter will read very close to zero volts, indicating the circuit or connection is not energized.
This adapter should not be used on low voltage control circuits or anywhere where the circuit under test could be adversely affected by this low impedance load. The adapter is designed to handle continuously applied power system voltages without damage, however proper use of this adapter is for intermittent use to determine whether a circuit is energized or not.
Sample measurement scenario when testing for stray electrical voltage:
This is the measurement displayed with the high impedance DMM between neutral and an unconnected wire in the same conduit as a 120 volt branch circuit feed. Note the high impedance meter is displaying 33 volts. This is a capacitively coupled stray voltage reading.
The last reading is displaying the result of the measurement from the previous reading when the stray voltage adapter is placed in the circuit. Note that the reading is now 13 millivolts or very close to zero volts, a non-energized connection.
Safety while testing for stray electrical voltage
On a more personal safety note to keep in mind as your testing for stray electrical voltage:
The Safest Way to Test Electrical Devices and Identify Electric Wires!The Non-Contact Electrical Tester
This is a testing tool that I have had in my personal electrical tool pouch for years, and is the first test tool I grab to help identify electrical wiring. It is a Non-contact tester that I use to easily Detect Voltage in Cables, Cords, Circuit Breakers, Lighting Fixtures, Switches, Outlets and Wires. Simply insert the end of the tester into an outlet, lamp socket, or hold the end of the tester against the wire you wish to test. Very handy and easy to use.
The Quickest Way to Check for Faulty Electrical Wiring!The Plug-In Outlet Tester
This is the first tool I grab to troubleshoot a problem with outlet circuit wiring. This popular tester is also used by most inspectors to test for power and check the polarity of circuit wiring.
It detects probable improper wiring conditions in standard 110-125 VAC outlets Provides 6 probable wiring conditions that are quick and easy to read for ultimate efficiency Lights indicate if wiring is correct and indicator light chart is included Tests standard 3-wire outlets UL Listed Light indicates if wiring is incorrect Very handy and easy to use.
Strip Off Wire Insulation without Nicking and Damaging the Electric Wire!The Wire Stripper and Wire Cutter
My absolute favorite wire stripping tool that I have had in my personal electrical tool pouch for years, and this is the tool I use to safely strip electrical wires.
This handy tool has multiple uses:
The wire gauges are shown on the side of the tool so you know which slot to use for stripping insulation.
The end of the tool can be used to grip and bend wire which is handy for attaching wire onto the screw terminals of switches and outlets..
The wire stripper will work on both solid and stranded wire. This tool is Very Handy and Easy to Use.
|Electrical Wiring Questions and Answers|